Digital Analog Tech Note 17 - TNDA17
PAXI0000, PAXI0000, PAXI0010, MPAXI000, MPAXI010
The RDSP 1 will revert to the value 339.2 even though another value has been programmed.
Cause of the Problem
When programming a PAXI for rate, if 0 segments are selected and the RDSP-1 value are programmed for a value of 65536 or higher, the PAXI RDSP 1 value will revert to 339.2. This is a function of PAXI firmware revision 2.0 only.
If RDSP-1 must be programmed to a value equal to or greater than 65536, program the meter to use one or more segments, not 0. This solution only results in two more pushes of the PAR button so it is simple to implement. Corrective Action Implementation: The next firmware version for the PAXI will correct this problem.
It is the customer's responsibility to review the advice provided herein and its applicability to the system. Red Lion makes no representation about specific knowledge of the customer's system or the specific performance of the system. Red Lion is not responsible for any damage to equipment or connected systems. The use of this document is at your own risk. Red Lion standard product warranty applies.
Red Lion Technical Support
If you have any questions or trouble contact Red Lion Technical Support by clicking here or calling 1-877-432-9908.
For more information: http://www.redlion.net/support/policies-statements/warranty-statement